Restore password × Upload manual upload from disk upload from url Thank you for your help! This error occurs when there is a problem with a table used by a driver function. Possible Cause: The instrument you are trying to communicate with is not at the expected primary address, the instrument is not powered on, or the cable to the instrument is either Device-level functions return ENEB when the specified access board of a device cannot be found in the system, even if the access board is configured in ibconf. 8 DMA hardware error http://glitchtest.org/error-code/gle-error-codes.html
This problem happens when the board is not physically plugged into the system, the I/O address specified during configuration does not mach the actual board setting, there is a system conflict If your GPIB board is the System Controller, then use the ibsic function (or SendIFC function) to send an interface clear. Check your system's resource manager to see if another board is trying to use part or all of this address range. This error is returned when the operation cannot be performed because of the existing lock on the interface.
This error might be caused by a failure to find or properly open the GPIB device driver. 1 Command requires GPIB Controller to be Controller-In-Charge. All rights reserved.| Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase English 7 ratings: 4.14 out of 5   GPIB Error Codes In case you are using the iblck command to lock the interface, increase the LockWaitTime and wait for the other process to relinquish control over the interface. My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
This error occurs when the driver is unable to wait for the RQS bit to set during a device-level ibwait. 17 Unrecognized command. 19 Board not present. 20 Table error. Don't show me this message again. All rights reserved. | Cart|Help You are here:NI Home > Support > Manuals > LabVIEW 2012 Help GPIB Error Codes »Table of Contents LabVIEW 2012 Help Edition Date: June Gpib Tutorial This error occurs when a driver function is called with an invalid parameter. 5 Command requires GPIB Controller to be System Controller.
For example, I may install a PCI-GPIB board in my computer and give a primary address of 2. Please try the request again. In this case, the shadow handshake is not possible and the error is returned to notify you of that fact. click to read more My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
FindRQS and AllSpoll return ETAB if the list of addresses to serial poll is empty or none of the devices in the list are asserting SRQ. 30 No GPIB address input. Gpib Commands If EOS is the termination method, then be sure to append the termination character to the end of your message. This error occurs while sending out GPIB command bytes. Often instruments will only respond to the standard commands after they have been placed in this mode, and otherwise they will give an error despite being sent a valid command.
This function is useful if you want your application to be notified asynchronously about the occurrence of one or more GPIB events. For legacy boards make sure that the jumpers and dip switches on the board are set to the same resource settings that the GPIB Configuration Utility thinks it is using. Gpib Configuration Utility Please try the request again. Gpib Interactive Control ESAC (5)Error Condition: GPIB board not System Controller as required.Description: ESAC results when ibsic, ibsre, SendIFC, or EnableRemote is called when the GPIB board does not have System Controller capability.
Byte count (where you expect to receive a certain number of bytes in a message) is always used, but some instruments use EOS and byte count, some use EOI and byte http://glitchtest.org/error-code/getlasterror-codes.html Verify that the cable is properly connected to the instrument. The default interface name for board is GPIB0, so it has a board index of 0, not 2.Solutions: Use the default settings for your GPIB hardware in the GPIB Configuration Utility Do not use a device descriptor in a board-level function or a board descriptor in a device-level function. Labview Error Code
EADR (3)Error Condition: GPIB board (GPIB0 or GPIB1) is not addressed correctly.Description: EADR occurs when the GPIB board is Controller-In-Charge (CIC) and is not properly addressing itself before read and write This error occurs when the board is the Controller-In-Charge and is not properly addressed before starting a transfer. Make sure that the interface name of your board is the same as the interface name of the board that your device is configured to communicate with (see KnowledgeBase 2368N85R, linked weblink This error may also occur, if the board index passed to the ibdev function is incorrect.
Then call ibnotify to re-enable notification EHDL(23)Error Condition: Input Handle is invalid.Possible Cause: Several GPIB commands take in the input handle of the board or the device as an input parameter Gpib-usb-hs Driver There are a few reasons why the instrument may not have anything to say: The message to the instrument may have been misspelled. If it does not appear, then you can call the ibwait function (set the wait mask for the CIC bit) to delay further processing until control is passed to your GPIB
Solutions: Give the GPIB board System Controller capability by calling ibrsc 1 or by using the GPIB Configuration Utility. Examine the variable before the failing function to make sure its value has not been corrupted. This error occurs when you try to take advantage of a driver feature that is either not implemented in the driver or is not currently usable. check over here ENOL (2)Error Condition: Function detected no Listener(s).Description: GPIB communications require a single Talker (to write data messages) and one or more Listeners (to read data messages).
Generated Thu, 24 Nov 2016 06:15:05 GMT by s_wx1196 (squid/3.5.20) ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://0.0.0.8/ Connection